Wuhan Dimed Laser Technology Co., Ltd
Dimedlaser - Proffessional Manufacture Of Medical Lasers
|Laser Type:||GaAlAs Diode Laser||Maximum Power:||15W/30W|
|Wavelength:||1210nm/980nm||Operation ModeCW, Single Or Repeat Pulse:||CW, Single Or Repeat Pulse|
Laser For Lipolysis
How the Cherylas 30watts 980nm works for lipolysis?
The procedures were performed in the hospital, and the type of anesthesia was general, regional (epidural), or local with sedation, according to the extent and indications of each case. using a 0.9% saline solution with a temperature of around 4°C, and 1:500,000 epinephrine. The infiltration was performed preferably with the aid of a continuous pressure system at 150 mm Hg, provided by placing the reservoir solution with a cuff pressure drop, to allow a homogeneous distribution during infusion with a 2-mm-diameter needle. A uniform deposit was performed only in the deepest layer, parallel to the muscular plane, to avoid tunneling the fat tissue. The infused volume was equivalent to the estimated volume drawn from each region. This infiltration technique took 7 to 8 min to infuse 1 liter into each decubitus, an ade -quate period for vasoconstriction.
A 980-nm-wavelength diode laser equipment was used in all treatments. A 2-mm-diameter rigid cannula with a 600-µm optical fiber was used to transfer energy to the tissues. The passage of the laser was initiated in the deep layers, with slow fan-shaped movements, at a rate of about 2.5 cm/s, from the incision, with application of back-and forth laser movements, always with the free control of the operator’s hand on the treated region to maintain the distance from the end of the fiber to the surface. At the end of the outward stroke of each passage, the fiber end was maintained at a safe distance from the skin, with the aid of a red light display, in order to avoid injury. After the passage of the annula into the deeper plane, passes were made with the same care in the middle plane, and then the more superficial plane to the skin surface.
For body treatment, power was maintained between 18 and 22 W, except in the submental region, inner arms, andthighs, where 14 to 16 W was used.
Now CHERYLAS 15 watts 1210nm wavelength has absorption-affinity for Lipid-Rich Tissue and has demonstrated to stimulate adipocytes and mesenchymal cells of the subcutaneous tissue, with a new concept: Selective Photothermostimulation (SPS). The application in Liposuction of this wavelength is determined by its preservation of the integrity of adipocytes, making possible the use of these materials for fat grafting in Liposculpture, Breast Reconstruction or as filler in Face Rejuvenation, proven that this technique is less traumatic.
Treatment areas include:•Chin / Jawline•Arms•Bra line•Abdomen•Hips•Buttocks•Thighs•Knees
The advantage of Laser Lipolysis:
|Laser Type||GaAlAs Diode Laser|
|Wavelength||810nm/ 980nm/ 1210nm/ 1470nm|
|Maximum Power||15W/ 30W/ 60W|
|Operation Mode||CW, Single Or Repeat Pulse|
|Pulse Duration||10us- 3s|
|Repetition Rate||0.2Hz- 50KHz|
|Pilot Beam||Red Diode Laser Of 650nm, Power<5mW|
|Control Mode||True Color Touch Screen|
|Transmission System||Medical Fibers With SMA905 Connector|
Laser lipolysis handpiece: